As an upgraded version of pH meter pro V1, the secondary generation pH meter pro greatly improves the precision and user experience. The onboard voltage regulator chip supports the wide voltage supply of 3.3~5.5V, which is compatible with 5V and 3.3V main control board, such as Arduino and LattePanda. The output signal filtered by hardware has low jitter. The software library adopts the two-point calibration method, and can automatically identify two standard buffer solutions (4.0 and 7.0), so simple and convenient. It uses an industry electrode and has built-in simple, convenient, practical connection and long life, very suitable for online monitoring.
This industry pH combination electrode is made of sensitive glass membrane with low impedance. It can be used in a variety of PH measurements with fast response, good thermal stability. It has good reproducibility, difficult to hydrolysis, and basically eliminate the alkali error. Between 0 to 14 pH range, the output voltage of the electrode is linear. The reference system which consist of the Ag/AgCl gel electrolyte salt bridge has a stable half-cell potential and excellent anti-pollution performance. The ring PTFE membrane is not easy to be clogged, so the electrode is suitable for long-term online detection.
With this product, the main control board (such as Arduino), and the software library, you can quickly build the pH meter, plug, and play, no welding. DFRobot provides a variety of water quality sensor products, uniform size, and interface, not only meet the needs of various water quality testing but also suitable for the DIY of multi-parameter water quality tester.
The pH is a value that measures the acidity or alkalinity of the solution. It is also called the hydrogen ion concentration index. The pH is a scale of hydrogen ion activity in solution. The pH has a wide range of uses in medicine, chemistry, and agriculture. Usually, the pH is a number between 0 to 14. Under the thermodynamic standard conditions, pH=7, which means the solution is neutral; pH7, which means the solution is alkaline.Attention：
1. The BNC connector and the signal conversion board must be kept dry and clean, otherwise, it will affect the input impedance, resulting in an inaccurate measurement. If it is damp, it needs to be dried.
2. The signal conversion board cannot be directly placed on a wet or semiconductor surface, otherwise, it will affect the input impedance, resulting in the inaccurate measurement. It is recommended to use the nylon pillar to fix the signal conversion board, allow a certain distance between the signal conversion board and the surface.
3. The sensitive glass bubble in the head of the pH probe should avoid touching with the hard material. Any damage or scratches will cause the electrode to fail.
4. After completing the measurement, disconnect the pH probe from the signal conversion board. The pH probe should not be connected to the signal conversion board without the power supply for a long time.Features
SpecificationSignal Conversion Board (Transmitter) V2
- 3.3~5.5V wide voltage input
- Hardware filtered output signal, low jitter
- Gravity connector and BNC connector, plug and play, no welding
- Software library supports two-point calibration, and automatically identifies standard buffer solution
- Uniform size and connector, convenient for the design of mechanical structures
- Supply Voltage: 3.3~5.5V
- Output Voltage: 0~3.0V
- Probe Connector: BNC
- Signal Connector: PH2.0-3P
- Measurement Accuracy: ±0.1℃
- Dimension: 42mm*32mm/1.66*1.26in
- Probe Type: Industrial Grade
- Detection Range: 0~14
- Temperature Range: 0~60°C
- Accuracy : ± 0.1pH (25 ℃)
- Response Time: 0.5 years (depending on the water quality)
- Cable Length: 500cm
- pH Probe (Industrial Grade) x1
- pH Signal Conversion Board V2 x1
- Gravity Analog Sensor Cable x1
- Waterproof Gasket x2
- Screw Cap for BNC Connector x1
- M3 * 10 nylon pillar x4
- M3 * 5 screws x8